- Hot search：
ISO 10343-2014 pdf free download
ISO 10343-2014 pdf free download.Ophthalmic instruments Ophthalmometers Instruments ophtalmiques-Ophtalmometres.
conversion value equal to 337,5 used to convert corneal curvature from inverse millimetres (mmi) to keratometric dioptres
[SOURCE: ISO 19980:2012,3.111
value of curvature, expressed in inverse millimetres (mm’), multiplied by the keratometric constant
[SOURCE: ISO 19980:2012, 3.12]
The ophthalmometer shall conform to the requirements specified in ISO 150044.
4.2 Radius of curvature measurements
The ophthalmonwter shall conform to requirements given in Table 1 or Table 2. Conformity shall be verified as described in 52. If the instrument conforms to the requirement of TahI1 it is a “Type B” ophthalmometer. If the instrument conforms to the requirements of Tablel it is a 9’ype A ophihalmometer. Type A and Type B ophthalmometers have sufficient resolution and range to adequately measure human corneas. Type A ophthalmometers have sufficient resolution and range (see Table 2) to adequately measure the radii of curvature of contact lenses complying with ISO 18369-3:2006,4.1.3, and Clause 5. It Is assumed that the local corneal front surface and both contact lens surfaces are spherical or toroidal.
4.4 Eyepiece adjustment (if applicable)
The dioptric adjustment range for distance-dependent Instruments shall be a minimum of-4 D to +4 D.
for which the scale from —3 0 to +2 D shall be calibrated.
5 Test methods
All tests described in this International Standard are type tests.
5.2 CheckIng optical requirements
Conformity to the requirements specified in 4,2 and 43 shall be verified by use of measuring devices the measuring error of which is less than 10 % of the smallest value to be determined.
Test results shall be evaluated according to the general rules of statistics.
Conformity to the requirements of 42 shall be verified using three spherical test surfaces, one chosen from each of three radii ranges: ≤6,8 mm, 7,5 mm to 8,1 mm and 9,1 mm. These test surfaces shall have the following properties:
a) uncertainty of sphere radius of curvature 1 pm;
b) local departure from sphericity 0.5 pm;ISO 10343-2014 pdf free download.
All resources on this website are from the Internet. It is only for personal study and research. No one is allowed to use it for any commercial purpose. If you need, please buy the genuine standard. If there is any infringement of your copyright, please contact the webmaster in time, and I will correct it immediately.
- Maybe you like
- Latest Updates
- ISO 287-2009 pdf free download
- ISO 527-5-2009 pdf free download
- ISO 659-2009 pdf free download
- ISO 660-2009 pdf free download
- ISO 712-2009 pdf free download
- ISO 725-2009 pdf free download
- ISO 730-2009 pdf free download
- ISO 927-2009 pdf free download
- ISO 2041-2009 pdf free download
- ISO 3040-2009 pdf free download
- ISO 3506-2-2009 pdf free download
- ISO 3506-3-2009 pdf free download