- Hot search：
ISO 21563-2013 pdf free download
ISO 21563-2013 pdf free download.Dentistry Hydrocolloid impression materials
Médecine bucco-dentaire — Produits pour empreintes a base d ‘hydro collo Ides.
c) Soft mouldable clay-like, putty-like or wax, to partially fill perforations in the ring mould [7.3.1 b)J so as to help keep the light bodied agar materials confined in the mould.
d) Flat glass specimen cover plate, approximately 50 mm x 50 mm and at least 3 mm thick having one side covered with a polyethylene sheet approximately 0,035 mm thick.
NOTE A very thin film of the high vacuum grease [7.1.1 b)] spread over the underside of the glass plate will keep the polyethylene sheet smoothly adapted to the plate.
e) Mould release solution (for the alginate containing specimens only), such as a freshly made 1 % solution of tetradecylamine in acetone, for preventing adherence of the alginate specimen materials to the lined test block surface.
1) Oven, held at (35 ± 2) °C, for simulated intra-oral temperature conditioning of the test block.
g) Equipment recommended for liquefaction, storing and tempering the agar materials and the recommended syringes and needles as may be applicable [8.2.2 a)] and the items needed for proportioning and mixing the alginates 18.2.3 c)J.
h) Two water baths, for use depending upon the material being tested
— one held within whatever temperature range the manufacturer instructions may recommend for cooling agar impressions in the mouth, or
— one held at simulated intra oral temperature of (35 ± 2) °C, for use with the alginate materials that are exposed to that approximate temperature while setting in the mouth.
I) C-clamp, having a minimum screw opening of 40 mm and a minimum throat depth of at least 30mm.
j) Disinfectant identified by the manufacturer [8.2.1 1)], second subdivisioni.
k) Microscope, equipped for x 4 to x 12 magnification and low angle illumination.
7.3.2 Examination and conditioning of equipment and accessories
a) Use the microscope [7.3.1 k)j to determine whether the lines on the test block have been cleared ot contaminants.
NOTE An ultrasonic cleaning device containing a concentrated solution of sodium bicarbonate in water has been found useful for removing alginate contaminants from impression trays and test accessories.
b) Place the test block [7.3.la)j in the oven 17.3.1 01 for conditioning at least 15 mm.
7.3.3 Specimen preparation — Agar materials
184.108.40.206 Prepare a minimum of three specimens.ISO 21563-2013 pdf free download.
All resources on this website are from the Internet. It is only for personal study and research. No one is allowed to use it for any commercial purpose. If you need, please buy the genuine standard. If there is any infringement of your copyright, please contact the webmaster in time, and I will correct it immediately.
- Maybe you like
- Latest Updates
- ISO 287-2009 pdf free download
- ISO 527-5-2009 pdf free download
- ISO 659-2009 pdf free download
- ISO 660-2009 pdf free download
- ISO 712-2009 pdf free download
- ISO 725-2009 pdf free download
- ISO 730-2009 pdf free download
- ISO 927-2009 pdf free download
- ISO 2041-2009 pdf free download
- ISO 3040-2009 pdf free download
- ISO 3506-2-2009 pdf free download
- ISO 3506-3-2009 pdf free download