- Hot search：
ISO 2476-2014 pdf free download
ISO 2476-2014 pdf free download.Butadiene rubber (BR) Solutionpolymerized types Evaluation procedures
Caoutchouc butadiène (BR) — Types polymérisés en solution — Méthodes d’évaluation.
5.2.2 Mixing procedures
Five mixing procedures are specified, a procedure using a laboratory internal mixer is preferred.
— method Al: single-stage mixing using a laboratory internal mixer;
— method A2: two-stage mixing using a laboratory internal mixer for both initial and final mixing;
— method B: two-stage mixing using a laboratory internal mixer for initial mixing and a mill for final mixing.
The smaller laboratory internal mixers do not provide enough compound for the final mill mixing since a batch mass of four times the formula mass is required. In these cases, the laboratory internal mixer should he used for the final mixing. The mixer head starting temperature or the batch mass can be adjusted, so that the final temperature of the discharged batch does not exceed 120 °C.
— method Cl and C2: mill mixing.
NOTE I These procedures can give different results.
NOTE 2 Mill mixing of solution-polymerized butadlene rubbers is more difficult than with other rubbers, and mixing is best accomplished by using an internal mixer. With some types of butadiene rubber, it is not possible to obtain a satisfactory mix using a mill-mixing procedure.
220.127.116.11 Method Al — Single-stage mixing using a laboratory internal mixer
For laboratory internal mixers having nominal capacities of 65 cm3 to about 2 000 cm3, the batch mass shall be equal to the nominal mixer capacity, In cubic centimetres. multiplied by the density of the compound. For each batch mixed, the laboratory internal mixer conditions shall be the same during the preparation of a series of identical mixes. At the beginning of each series of test mixes, a machine- conditioning batch shall be mixed using the same formulation as the mixes under test. The laboratory internal mixer shall be allowed to cool down to a specified temperature (60 °C has been found to be suitable) between the end of one test batch and the start of the next. The temperature control conditions shall not be altered during the mixing of a series of test batches.
The mixing technique shall be such as to obtain a good dispersion of all the ingredients.
The temperature of the hatch discharged on completion of mixing shall not exceed 120 °C. If necessary, adjust the batch mass or the mixer head starting temperature so that this condition is met.
NOTE 1 The mixing conditions given In Table A.7 for various sizes of laboratory internal mixer might be helpful. NOTE 2 A general mixing procedure for the laboratory internal mixer is as follows:ISO 2476-2014 pdf free download.
All resources on this website are from the Internet. It is only for personal study and research. No one is allowed to use it for any commercial purpose. If you need, please buy the genuine standard. If there is any infringement of your copyright, please contact the webmaster in time, and I will correct it immediately.
- Maybe you like
- Latest Updates
- ISO 287-2009 pdf free download
- ISO 527-5-2009 pdf free download
- ISO 659-2009 pdf free download
- ISO 660-2009 pdf free download
- ISO 712-2009 pdf free download
- ISO 725-2009 pdf free download
- ISO 730-2009 pdf free download
- ISO 927-2009 pdf free download
- ISO 2041-2009 pdf free download
- ISO 3040-2009 pdf free download
- ISO 3506-2-2009 pdf free download
- ISO 3506-3-2009 pdf free download