- Hot search：
ISO 34-1-2015 pdf free download
ISO 34-1-2015 pdf free download.Rubber, vulcanized or thermoplastic Determination of tear strength
Part 1:Trouser, angle and crescent test pieces
Caoutchouc vulcanisé ou thermoplastique — Determination de la résistance a u déch irem en t — Partie 1: Eprouvettes pan talon, angulaire et croissant.
5.2 Nick cutter
A sharp razor blade or a sharp knife free from ragged edges shall be used for producing a cut or a nick in the test piece.
The apparatus for introducing the nick required for the nicked angle or crescent test piece shall be as follows.
Means shall be provided for clamping the test piece firmly, especially in the region where the nick is to be introduced. The cutting tool, consisting of a razor blade or similar blade, shall be clamped in a plane perpendicular to the major axis of the test piece, and positioned so as to introduce the nick in the appropriate place. The blade clamping device shall permit no lateral movement and shall be fitted in guides to enable the blade to be moved across the test piece with its edge remaining perpendicular to the plane of the test piece. Alternatively, the blade shall be fixed and the test piece arranged to move in an analogous manner. Means shall be provided for fine adjustment of the depth of the nick. The adjustment of the position of the blade holder or clamped test piece shall be determined for each blade by cutting one or two preliminary nicks and measuring these with the aid of a microscope. The blade shall be wetted with water or soap solution prior to nicking.
NOTE A suitable apparatus for nicking tear test pieces has been described in detail In the literature.161
To check that the depth of the nick is within the specified limits (see Z4) any suitable means may be used, e.g. an optical projection apparatus, A convenient arrangement isa microscope giving at least lOx magnification fitted with a travelling stage suitably illuminated. The eyepiece is fitted with a graticule or crosswire by which to record the travel of the stage and test piece through a distance equal to the depth of the nick. The travel of the stage is calibrated with a stage micrometer.
Alternatively, a travelling microscope may be used.
The apparatus shall have an accuracy of measurement of 0,05 mm.
5.3 TestIng machine
The machine shall conform to the requirements of ISO 5893, to an accuracy corresponding to class 1.
It shall be capable of registering the applied forces within I % during the test while maintaining the specified constant rate of separation of the jaws of 100 mm/mm ± 10 mm/mm for the trouser test piece and 500 mm/mm ± 50 mm/mm for the angle and crescent test pieces. A low-inertia machine having autographic force-recording facilities is essential when using the trouser test piece.
The machine shall be provided with a type of grip which tightens automatically as the tension increases and exerts a uniform pressure across the widened end of the test piece.ISO 34-1-2015 pdf free download.
All resources on this website are from the Internet. It is only for personal study and research. No one is allowed to use it for any commercial purpose. If you need, please buy the genuine standard. If there is any infringement of your copyright, please contact the webmaster in time, and I will correct it immediately.
- Maybe you like
- Latest Updates
- ISO 287-2009 pdf free download
- ISO 527-5-2009 pdf free download
- ISO 659-2009 pdf free download
- ISO 660-2009 pdf free download
- ISO 712-2009 pdf free download
- ISO 725-2009 pdf free download
- ISO 730-2009 pdf free download
- ISO 927-2009 pdf free download
- ISO 2041-2009 pdf free download
- ISO 3040-2009 pdf free download
- ISO 3506-2-2009 pdf free download
- ISO 3506-3-2009 pdf free download