- Hot search：
AWS C3.5M/C3.5:2016 pdf free download
AWS C3.5M/C3.5:2016 pdf free download.Specification for Induction Brazing.
and ANSI/NCSL Z540.3, Requirements fi.r (lie calibration of Measuring and Test Equijnnent. Instruments shall be calibrated on a regular schedule or whenever repairs or modifications are made to them.
For joints in which the overheating of the base metals could adversely affect the properties of the base metal, sensors that shut down the system shall be used if overheating occurs and shall be calibrated as above. Each sensor shall be set at a temperature below the temperature at which the reduction in properties can take place. It shall be located and aligned so that it reads the area of highest temperature. Automatic systems used to manufacture Class A and Class B joints shall be equipped with calibrated control devices that produce a repeatable brazing cycle.
188.8.131.52 To prevent interaction between the induction field and thermocouple-type temperature sensors, noncontact temperature sensing devices may be used to monitor temperature. If used, these devices should be used with caution as they are greatly influenced by the condition of the surface being measured. Flux, oxidation, and material ernissivity will affect the infrared signature of the material.
5.2.5 Atmosphere Induction Brazing Equipment. If induction brazing is to he conducted in a protective atmosphere (including vacuum), the equipment and enclosures shall he in good working condition and shall he as specified in the qualified brazing procedure specification (see 5.4 and 5.5).
184.108.40.206 Electronic Dew Point Measuring Devices. Electronic dew point measuring devices, when used, shall be calibrated in accordance with the manufacturer’s recommendations against reference devices traceable to the National Institute of Standards and Technology (NIST), or equivalent. Such gauges shall be calibrated in accordance with NCSL Z540.3 before initial use and recalibrated at a frequency not to exceed six months.
220.127.116.11 Vacuum Measuring Devices. When used, thermocouple gauges shall be used to measure pressure higher than 0.13 Pa [10—3 torn. Suitable electronic gauges shall he used to measure pressures of 0.13 Pa [10—3 torn or less. These gauges shall he so mounted as to measure the pressure within the work zone of the chamber. Suitable instruments for reading and recording the pressure within the chamber work zone shall also he provided.
These instruments and gauges shall be calibrated in accordance with the manufacturer’s recommendations against reference standards traceable to NIST. Such gauges shall be calibrated in accordance with NCSL Z540.3 before initial use and recalibrated at a frequency not to exceed six months.AWS C3.5M/C3.5:2016 pdf free download.
All resources on this website are from the Internet. It is only for personal study and research. No one is allowed to use it for any commercial purpose. If you need, please buy the genuine standard. If there is any infringement of your copyright, please contact the webmaster in time, and I will correct it immediately.
- Maybe you like
- Latest Updates
- ISO 287-2009 pdf free download
- ISO 527-5-2009 pdf free download
- ISO 659-2009 pdf free download
- ISO 660-2009 pdf free download
- ISO 712-2009 pdf free download
- ISO 725-2009 pdf free download
- ISO 730-2009 pdf free download
- ISO 927-2009 pdf free download
- ISO 2041-2009 pdf free download
- ISO 3040-2009 pdf free download
- ISO 3506-2-2009 pdf free download
- ISO 3506-3-2009 pdf free download