- Hot search：
IEEE 644-2019 pdf free download
IEEE 644-2019 pdf free download.IEEE Standard for Measurement of Power Frequency Electric and Magnetic Fields from AC Power Lines.
3.1 General characteristics
Three types of meters used to measure the electric field strength from ac power lines are described in technical
— Free-body meter: Measures the steady-state induced current or charge oscillating between two halves of an isolated conductive body in an electric field (see Adler [BI 1 and Bracken [B2]).
– Ground-reference-type meter: Measures the current-to-ground from a flat probe introduced into an electric field (see Miller ).
Electro-optic field meter: Measures changes in the transmission of light through a fiber or crystal
due to the influence of the electric field (see Cooke [B4j and Hamasaki ).
The free-body meter is suitable for survey-type measurements because it is portable, allows measurements above the ground plane, and does not require a known ground reference. Therefore, this type of meter is recommended for outdoor measurements near power lines.
This standard presents measurement techniques for only the free-body type meter. Flat ground-reference- type meters can be used only under special conditions described in 5.2. Electric field strength meters intended fhr characterization of radio-frequency electric fields should not be used to measure the power- frequency electric field strength from ac power lines.
An electric field strength meter consists of two primary parts, the probe and the detector. For commercially available free-body meters. the detector is usually contained in, or is an integral part of. the probe. The probe and detector arc introduced into an electric field on an insulating handle. The detector measures the steady-state induced current or charge oscillating between the conducting halves (electrodes) of the probe. The observer is sufficiently removed from the probe to avoid significant perturbation of the electric field at the probe (see 4.1). The size of the probe should be such that charge distributions on the boundary surfaces generating the electric field (energized and ground surfaces) are, at most, weakly perturbed when the probe is introduced for measurement. The electric field should be approximately uniform in the region where the probe will be introduced. Probes can be of any shape; however, meters commercially available in the U.S. are generally in the shape of rectangular boxes, with side dimensions ranging from approximately 7 cm to 20 cm. The meters are calibrated to read the rms value of the power frequency electric field component along the electrical axis (the axis of greatest electric field strength sensitivity).IEEE 644-2019 pdf free download.
All resources on this website are from the Internet. It is only for personal study and research. No one is allowed to use it for any commercial purpose. If you need, please buy the genuine standard. If there is any infringement of your copyright, please contact the webmaster in time, and I will correct it immediately.
- Maybe you like
- Latest Updates
- ISO 287-2009 pdf free download
- ISO 527-5-2009 pdf free download
- ISO 659-2009 pdf free download
- ISO 660-2009 pdf free download
- ISO 712-2009 pdf free download
- ISO 725-2009 pdf free download
- ISO 730-2009 pdf free download
- ISO 927-2009 pdf free download
- ISO 2041-2009 pdf free download
- ISO 3040-2009 pdf free download
- ISO 3506-2-2009 pdf free download
- ISO 3506-3-2009 pdf free download