- Hot search：
BS 8494-2007 pdf free download
BS 8494-2007 pdf free download.Electronic portable and transportable apparatus designed to detect and measure carbon dioxide in indoor ambient air — Requirements and test methods.
22.214.171.124 The resistance to impact of the apparatus shall be tested in accordance with BS EN 60079-0:2004, 26.4.2, and be capable of withstanding a low impact resistance for Group II apparatus in accordance with BS EN 60079-0:2004, Table 8. If mechanical protection such as a carrying case is provided, this test shall be carried out with the apparatus assembled in the form in which it is intended to be used.
126.96.36.199 Adjustments that might affect the performance of the apparatus shall be suitably protected against unauthorized changes, such as through the use of passwords, tamper-evident seals or special tools.
4.1.3 Power supplies
188.8.131.52 For battery-powered apparatus, the batteries specified by the manufacturer shall have sufficient capacity to enable accurate readings of gas concentration and other measurement parameters, if provided, for at least 4 Ii of continuous usage. ‘When the battery capacity falLs below a preset level a visual warning shaLl be given, following which the apparatus shall operate to the performance specification for at least a further :30 nun.
184.108.40.206 Apparatus provided with the means of connection to the mains supply shall conform to relevant British Standards.
NOTE This u,u is coer& by iarious slot utury regithz!ims.
220.127.116.11 Indicators shall be provide(l to show:
a) mea.sure(l parameter, values and units of measurement:
5) low battery warning for I)attery-powered apparatus;
c) for aspirated apparatus with an integrate(l (lOW indicator, a warning when a blockage occurs (see 6.3.13);
d) mode of operation or apparatus status (e.g. warm-up and uwasuirement); and
e) when values lie outside the indicating range.
18.104.22.168 All values shall be indicated on a display with characters not less than 8 mm high, unless adequate provision is made for enhancing the legibility (e.g. backlighting), in which case the minimum character height shall be 4 mm.BS 8494-2007 pdf free download.
All resources on this website are from the Internet. It is only for personal study and research. No one is allowed to use it for any commercial purpose. If you need, please buy the genuine standard. If there is any infringement of your copyright, please contact the webmaster in time, and I will correct it immediately.
- Maybe you like
- Latest Updates
- ISO 287-2009 pdf free download
- ISO 527-5-2009 pdf free download
- ISO 659-2009 pdf free download
- ISO 660-2009 pdf free download
- ISO 712-2009 pdf free download
- ISO 725-2009 pdf free download
- ISO 730-2009 pdf free download
- ISO 927-2009 pdf free download
- ISO 2041-2009 pdf free download
- ISO 3040-2009 pdf free download
- ISO 3506-2-2009 pdf free download
- ISO 3506-3-2009 pdf free download