- Hot search：
UL 2808-2020 pdf free download
UL 2808-2020 pdf free download.Energy Monitoring Equipment.
19.2 After 30 minutes, the current sensor shall be removed from the tank, the excess water removed from the surface of the case, and the Dielectric Voltage Withstand Test, Section j4. shall be conducted. After the test, the sensor case shall be opened and a visual inspection made,
19.3 The current sensor shall comply with the requirements of the Dielectric Voltage Withstand Test, Section 14. and the visual inspection confirms no water has entered an area of the case that has exposed live parts that are not covered by water resistant insulation or at least 0.8 mm (0.03 in) of potting material that complies with J,Q.
20 Outdoor Use Cold Impact Test
20.1 All enclosed-type current sensors shall be subjected to the test described in this Section.
20.2 Three representative specimens of each current sensor series rated for cold ambient use shall be cooled to the manufacturer’s defined low ambient rating, i2.0’C (3.6F). Enclosed-type current sensors shall be cooled to the above, or minus 35.0 ±2,0 C (-31 ±3.6’ F), whichever is colder. Sensor5 shall be maintained at this temperature for at least 3 hours.
20.3 While the unit is still cold, each specimen shall be subjected to the impact of 6.8 J (5 foot-pounds). produced by dropping a steel sphere 50.8 mm (2 in) in diameter and weighing 0.54 kg (1.18 lbs) from a height of 1.3 m (51 in). or by swinging the steel sphere as a pendulum from a height of 1.3 m (51 in).
20.4 Compliance is visually determined. There shall be no damage that would allow access to live parts, that reduce required spacings. or for enclosed-type current sensors, would allow the ingress of water to live parts.
CURRENT SENSOR MARKINGS
Advisory Note: In Canada. there are two official languages. English and French. Annex provides translations in French of the English safety markings specified in this standard. Markings required by this standard may have to be provided in other languages to conform with the language requirements of the country where the product is to be used.
211 The markings. including signage required for compliance to this standard, shall be legible and permanent such as metal stamped. etched, adhesive labels, etc. An adhesive-backed label shall comply with UL 969. and CSA C22.2 No. 0.15, for the intended exposure conditions and surface adhered to.
21.2 Each current sensor shall be marked with the following:UL 2808-2020 pdf free download.
All resources on this website are from the Internet. It is only for personal study and research. No one is allowed to use it for any commercial purpose. If you need, please buy the genuine standard. If there is any infringement of your copyright, please contact the webmaster in time, and I will correct it immediately.
- Maybe you like
- Latest Updates
- ISO 287-2009 pdf free download
- ISO 527-5-2009 pdf free download
- ISO 659-2009 pdf free download
- ISO 660-2009 pdf free download
- ISO 712-2009 pdf free download
- ISO 725-2009 pdf free download
- ISO 730-2009 pdf free download
- ISO 927-2009 pdf free download
- ISO 2041-2009 pdf free download
- ISO 3040-2009 pdf free download
- ISO 3506-2-2009 pdf free download
- ISO 3506-3-2009 pdf free download