- Hot search：
UL 2992-2018 pdf free download
UL 2992-2018 pdf free download.Environmental Claim Validation Procedure (ECVP) for Determination and Verification of Chemical Emissions from Water Activated Flameless Heat Packets.
4.1 For the purpose of this procedure, the following definitions apply.
4.2 Air Exchange Rate (ACH) — Air change rate is a measure of the air volume added to or removed from a space divided by the volume of the space.
4.3 Air Flow Rate — Air volume entering the emission test chamber per unit time.
4.4 Background Concentrations — Airborne chemical or particle level in the emission test chamber in the absence of a product test specimen.
4.5 Emission Rate (ER) — The rate of emission of a specific analyte defined as the total rig/hr of a chemical emitted from a product.
COMPLIANCE, EVALUATION AND ASSESSMENT CRITERIA
5 Section I: Testing Procedure
5.1 Facility and equipment requirements
511 Facility requirements
18.104.22.168 Facilities designed and operated to measure chemical emissions from water activated flameless heat packets during exothermic reaction should contain emission test chambers, air purification systems. monitoring and control systems. sample collection and analysis equipment. standards generation and calibration systems. data acquisition systems. and data modeling and report systems. Facilities should demonstrate compliance with Standard for General Requirements for the Competence of Testing and Calibration Laboratories. ISO 17025, with the referenced test method included in the scope.
5.1.2 Emission test chamber requirements
22.214.171.124 The emission test chamber volume shall be greater than or equal to 5 m3
126.96.36.199 Test chambers with a different volume may be accepted provided that sufficient data support evidence of comparable results.
188.8.131.52 The emission test chamber shall be operated according to the guidelines of Standard Practice for
Full-Scale Chamber Determination of Volatile Organic Emissions from Indoor Malerials/Products, ASTM
D6670. using inlet air at 23 ±2C and 50 ±5% relative humidity with 1 air changes per hour.
184.108.40.206 Teflon tubing shall be inserted through a port in the chamber to facilitate adding water to the heat packets.UL 2992-2018 pdf free download.
All resources on this website are from the Internet. It is only for personal study and research. No one is allowed to use it for any commercial purpose. If you need, please buy the genuine standard. If there is any infringement of your copyright, please contact the webmaster in time, and I will correct it immediately.
- Maybe you like
- Latest Updates
- ISO 287-2009 pdf free download
- ISO 527-5-2009 pdf free download
- ISO 659-2009 pdf free download
- ISO 660-2009 pdf free download
- ISO 712-2009 pdf free download
- ISO 725-2009 pdf free download
- ISO 730-2009 pdf free download
- ISO 927-2009 pdf free download
- ISO 2041-2009 pdf free download
- ISO 3040-2009 pdf free download
- ISO 3506-2-2009 pdf free download
- ISO 3506-3-2009 pdf free download