- Hot search：
UL 82-2018 pdf free download
UL 82-2018 pdf free download.Electric Gardening Appliances.
Exception: Only a single sample of a gardening appliance employing an alternate magnet wire coating is to be tested lithe test results are equivalent to the test results of the original construction tested. Equivalency is based upon the test duration, the enclosure temperature, and the leakage current being slightly higher. the same, or less and acceptable dielectric voltage withstand test results.
27 Leakage Current Test
27.1 An appliance rated for a nominal 120volt supply, shall be subject to the test specified in 27.2 —27.7. As a result of the test, the leakage current shall not be more than 0.5 milliamperes.
27.2 All exposed conductive surfaces are to be tested for leakage currents. The leakage currents from these surfaces are to be measured to the grounded supply conductor individually as well as collectively where simultaneously accessible. Parts are considered to be exposed surfaces unless guarded by an enclosure considered acceptable for reducing the risk of electric shock as defined in 10.1.1 — 10.1.6. Surfaces are considered to be simultaneously accessible if they can be readily contacted by one or both hands of a person at the same time. These measurements do not apply to terminals operating at voltages that are considered not to present a risk of electric shock.
27.3 It a conductive surface other than metal is used for the enclosure or part of the enclosure, the leakage current is to be measured using a metal foil with an area of 10 by 20 centimeters (3.9 by 7.9 inches) in contact with the surface. If the surface is less than 10 by 20 centimeters (3.9 by 7.9 inches), the metal foil is to be the same size as the surface. The metal toil is not to remain in place long enough to affect the temperature of the appliance.
27.4 The measurement circuit for leakage current is to be as illustrated in Figure 27.1. The measurement instrument is defined in (a) — (c). The meter that is actually used for a measurement need only indicate the same numerical value for a particular measurement as would the defined instrument. The meter used is not required to have all the attributes of the defined instrument.
a) The meter is to have an input impedance of 1500 ohms resistive shunted by a capacitance of 0.15 microfarad.
b) The meter is to indicate 1.11 times the average of the full-wave rectified composite wave
form of voltage across the resistor or current through the resistor.
C) Over a frequency range of 0 — 100 kIlohertz, the measurement circuitry is to have a
frequency response ratio of indicated to actual value of current that is equal to (he ratio of the impedance of a 1500 ohm resistor shunted by a 0.15 microfarad capacitor to 1500 ohms. At an indication of 0.5 milliampere the measurement is to have an error of not more than 5 percent at 60 hertz.UL 82-2018 pdf free download.
All resources on this website are from the Internet. It is only for personal study and research. No one is allowed to use it for any commercial purpose. If you need, please buy the genuine standard. If there is any infringement of your copyright, please contact the webmaster in time, and I will correct it immediately.
- Maybe you like
- Latest Updates
- ISO 287-2009 pdf free download
- ISO 527-5-2009 pdf free download
- ISO 659-2009 pdf free download
- ISO 660-2009 pdf free download
- ISO 712-2009 pdf free download
- ISO 725-2009 pdf free download
- ISO 730-2009 pdf free download
- ISO 927-2009 pdf free download
- ISO 2041-2009 pdf free download
- ISO 3040-2009 pdf free download
- ISO 3506-2-2009 pdf free download
- ISO 3506-3-2009 pdf free download